Proteus Industries presents; Comparison of two Coriolis meters from +5⁰C to +120⁰C fluid temperature at the European Flow Measurement Conference in Barcelona
Coriolis Calibrations are done on Flow, Temperature, and Density. They operate in wide temperature ranges however they are not calibrated outside of +5C and +90C fluid temperatures.
Common industry practice involves using measurement uncertainty analysis and curve projections to estimate the meter’s performance and behavior in such conditions. In this study, two recently calibrated Coriolis Meters were compared and analyzed from +25C to +120C.
The collected data was then compared to the calculated uncertainty error for each Coriolis Meter. The results showed that a water-based calibration above +5C and +90C can be done with proper procedure and temperature controlled process.